Force = mass (Kilogram) x Acceleration (metre per second squared)
The international prototype of the kilogram, an artefact made of platinum-iridium, is kept at the BIPM under the conditions specified by the 1st CGPM in 1889 when it sanctioned the prototype and declared.
The kilogram is the unit of mass; it is equal to the mass of the international prototype of the kilogram.
National Metrolpgy Institute (NMI)
In the United States the National Institute of Standards and Technology (N.I.S.T.) is the designated NMI
N.I.S.T. uses dead weight primary standards known to within 0.0004 % of applied up to 4.4 MN or 1,000,000 LBF
Primary Reference Laboratories
Primary Reference Labs use dead weight Primary Standards, usually calibrated by N.I.S.T. as the reference standard. Morehouse Instrument Company uses dead weight primary standards calibrated directly by N.I.S.T up to 533 kN or 120,000 LBF known to within 0.008 % of applied.
Morehouse Calibration and Measuring Capabilities (CMC) as reported on our scope of accreditation is 0.0016 % to 0.002 % of applied using dead weight primary standards. (Note 0.0016 % is for K=2)
Accredited Calibration Service Supplier
Accredited Calibration Suppliers use secondary standard (those calibrated by primary standards) Typically these standards are capable of measuring forces to within 0.02 % of applied typically for k=1. When working with several standards such as ASTM E74, ISO 376 and others, these labs are often required to have their instrumentation calibrated by dead weight primary standards. If you are not working to ASTM, ISO 376 or other standards requiring calibration dead weight primary standards you will need to determine the the appropriate uncertainty needed to calibrate other Measuring and Test Equipment (MT&E)
Working standards and certified reference materials used in commerce and industry have a traceable relationship to the secondary and primary standards. This is the category where you need to determine the the appropriate uncertainty needed to calibrate your Measuring and Test Equipment (MT&E). If you are accredited and using dynamometers, force gauges, load cells or other MT&E to make critical measurements, you will want to use a calibration laboratory with Calibration and Measuring Capabilities low enough so that the Probability of False Accept (PFA) is low enough to ensure you are using these devices within the stated tolerances.
Note: The risk of incorrectly declaring a device as in-tolerance (false-accept risk) is dependent upon several factors. Those factors include the specified tolerance limit, guard-band, the calibration process uncertainty etc..,
These are typically the material testing machines used for tensile, compression, fatigue, impact, rheology, structural, materials or a range of other force applications.
What does this mean as far as the Uncertainty of Your Standards?
Your Measurement Uncertainty is directly affected by the standards used to perform the calibration.
When you calculate measurement uncertainties in accordance with ISO “Guide to the Expression of Uncertainty in Measurement” (GUM)1, A2LA R205, and ILAC P-14 as required by ISO/IEC 17025 guidelines you will need to consider the following:
Repeatability (Type A)
Reference Standard Uncertainty
Reference Standard Stability
You can use these standard contributors to derive your CMC (Calibration and Measurement Capability) which is then reported to your customer, among other things
Let's examine CMC and what the reference lab's CMC does to the calibration results
Primary Calibration Laboratory CMC
Morehouse Primary Standard as the Reference (CMC 0.0016 % for K=2 or 0.16 LBF @ 10K)
Morehouse CMC is 14.43 % of the Total Expanded Uncertainty (U)
Expanded Uncertainty @ 10K = 0.41 LBF
Morehouse CMC = 0.16 LBF Repeatability = 0.057 LBF
Accredited Calibration Supplier
Accredited Calibration Supplier with Secondary Standards as the Reference (CMC 0.04 % for K=2 or 4 LBF)
Accredited Calibration Supplier 95.74 % of the Total Expanded Uncertainty (U)
Expanded Uncertainty @ 10K = 4.03 LBF
Accredited Cal Supplier CMC = 4.00 LBF, Repeatability = 0.379 LBF
When selecting a calibration laboratory it is important to evaluate your tolerances and the Calibration and Measurement Capabilities (CMC's) of your calibration supplier. Lower uncertainties typically equate to more accurate measurements, less risk, long term cost savings and consumer confidence.
Written by Henry Zumbrun